Views:4 Author:Site Editor Publish Time: 2020-05-26 Origin:Site
How to use shaking incubator correctly?
The shaking incubator / shaker is a commonly used as laboratory equipment, which is widely used in bacterial cultivation, fermentation, hybridization, biochemical reaction, enzyme and tissue research, etc., which have high requirements on temperature and oscillation frequency. At the same time, it is also indispensable laboratory equipment for cultivation and preparation in scientific research, education and production departments such as plants, biology, microbes, medicine, environmental protection and so on. Do you know how to use the shaking incubator correctly?
Laboratory operation instructions
When you use the laboratoryshaking incubator / shaker, please follow the instructions before, during and after installation to avoid unnecessary failure of the equipment. In order to operate the device, please read the instruction manual in detail and follow the requirements in the following aspects:
Place the shaking incubator / shaker on a horizontal surface or workbench that can support the weight of the shaker and all related parts.
2. Operating environment:
Please ensure that the indoor temperature range is between 18 ℃ ~ 34 ℃ and the relative humidity is up to 90%.
3. Power supply voltage:
Please confirm the local power supply voltage meets the requirements before use.
4. Component installation:
The installation of the shaking incubator clamp is to fix the base of the clamp to the platform by using the correct type and number of screws.
5. Standard operation:
When the shaking incubator / shaker is rotating at high speed, in order to avoid large vibration of the instrument, the culture reagent bottles should be placed symmetrically on the shaker, and the culture medium in each bottle should be approximately equal. Please fix the sample firmly on the shaker. When the shaker is working, the sample will not be shaken violently before leaving. During the sampling process, if there is a sample missing (including the drop of the plug), the main power supply should be turned off, as far as possible to remove debris with tweezers. If necessary, some parts need to be disassembled for cleaning, do not directly handle by hand when the shaker is working. It is forbidden to put the tube rack made of plastic foam into the shaker, so as to avoid foam particles sucked into the fan shaft, causing the fan to overheat and burn! When removing the sample, turn off the power or set the speed to zero.
The surface of the shaking incubator / shaker should be scrubbed frequently to keep it clean. When scrubbing, the side of the instrument control box should be avoided to prevent water from entering the instrument and damaging the instrument. During regular use, regularly check whether the fastening screws of the fuse box, control components and instruments are loose. For long-term use, when the instrument fails, it should be repaired by professional maintenance personnel and cannot be easily disassembled. When it is not used for a long time, especially in the rainy or humid season, it should be regularly energized and operated for 5-8 hours to drive off the moisture absorbed by the electrical components of the equipment.
Zhichu is a company that has been devoted to R & D, production, sales and service of shaking incubator / shaker for many years. The production equipment, the strong technical force, the standard process flow, and the unique management mode make Zhichu shaking Incubator / shaker become a rising star in the industry.