Views: 3 Author: Site Editor Publish Time: 2021-06-07 Origin: Site
1. The purpose of the illumination shaking incubator:
The illumination shaking incubator adds the illumination function to the shaking incubator. The instrument can set different working temperatures and different light intensities according to different times, so that plants can achieve experimental results in a short time. The light shaking incubator is equipped with an over-temperature abnormal protection function to ensure the safety of the instrument's samples. Optional full-spectrum plant growth lights are conducive to the growth of plants and improve disease resistance. It has the functions of power-down memory and power-off time automatic compensation; constant temperature control system, quick response and high temperature control accuracy.
2. Features of the illumination shaking incubator:
The illumination shaking incubator adopts a full microcomputer automatic control system, programmable multi-segment program control mode, and can set up to 99 time periods of different temperature, humidity and illuminance parameters every day. Forced convection air duct ventilation, the studio wind speed is soft, the temperature is uniform; hollow reflective tempered coated glass, good heat insulation performance, beautiful and generous.
3. Technical parameters of the illumination shaking incubator:
1. Volume: 1L*9 、1L*24、 1L*84
2. Temperature control range: 4~60℃ (at room temperature 23℃~25℃)
3. Temperature uniformity: ±1℃
4. Light intensity: 0~16000LX or 0~30000LX (infinitely adjustable)
5. Timing range: 0~999.9 hours
Fourth, the maintenance of the illumination shaking incubator：
1. The shell of the incubator should be grounded reliably.
2. The incubator should be placed in a dry, cool, well-ventilated place away from the sun and heat sources. Place it steadily to avoid vibration and noise.
3. To ensure effective heat dissipation of the condenser, the distance between the back of the box and the wall should be greater than 10cm. There should be a 5cm gap on the side of the box and at least 30cm on the top of the box.
4. When repairing, transporting and maintaining the incubator, avoid collision, shaking and vibration; the inclination should be less than 45 degrees.
5. The instrument suddenly stops working. Please check whether the fuse (rear side of the cabinet) is burned out if the power supply is normal.
6. When the incubator is in refrigeration, it is not advisable to make the difference between the temperature in the incubator and the ambient temperature more than 25 degrees.
Zhichu light shaking incubator integrates a full temperature shaking incubator and a caring incubator, which can be used for static cultivation or dynamic cultivation, one machine for two purposes. Special LED lamp design, high efficiency and energy saving, high light efficiency, light intensity up to 30000LX, infinitely adjustable, long service life.